The Stainless Steel Malvern 3000 Enclosure with Getinge La Calhene Alpha- Rapid Transfer Port (RTP) port was designed to provide personnel protection during spectrophotometric analysis of Highly Potent Pharmaceutical Ingredient (HPAPI) powder. The enclosure features a 316 grade stainless steel superstructure and a hinged front door with gas springs for loading and unloading of equipment and other materials.
On the right side, the enclosure features (2x) NEMA-rated receptacles, (2x) 3” diameter iris ports for electrical connections, and a control box with FS1650 integrated alarm. Also features two 3/8” NPT ball valves with turret for connection to gas sources (½” barb fitting angled 90 degrees). The enclosure workspace allows the operator to efficiently load samples into the Hydro SV cuvette and insert the unit into the Mastersizer 3000 optical bench. The workspace accommodates sufficient space for a Hydro SV with dimensions 4.5”(L) x 11”(W) x 8.5”(H) and 27”(W) x 12”(D) x 18” (H) for the Malvern Panalytical Mastersizer 3000. Laminated safety glass viewing panels and LED lighting provide optimal field of vision and illumination across the workspace.
40” Exterior Width
44” Exterior Depth
36” Interior Height
FSI Category: 10+
Performance is paramount in the safety industry. Through consistent design quality and expert manufacturing, Flow Sciences’ units perform. With surrogate powder testing both in our facility as factory acceptance testing (FAT) and at the customer facility as site acceptance testing (SAT), we consistently exceed our customers’ containment targets and goal expectations.
FLOW SCIENCES CONTAINS. FLOW SCIENCES PERFORMS.
Containment Target : 100 ng/m³
Surrogate Powder Testing Result : 0.06 ng/m³
Equipment : Balance
Operation : Sympatec Simulation, Cleaning
Test : Naproxen Sodium – 3 x 100g